ZYTTRIA® products are not sterile on delivery.
Sintered prosthetic structures can be sterilised with EO gas(at a gas concentration of 900mg/L for 3 hours at 55°C) with use of EO gas sterilizer (of the kind of HS-700 EO, HANSHIN-Korea).
Healthcare professional is responsible for the sterility of dental prosthesis, ensuring use and control of equipments and products suitable for cleaning, disinfection and sterilisation, in addition to correct procedures, following legal regulations.
Validation of method of sterilisation is charged to certified healthcare professionals.
Attention! Do not sterilise in an autoclave or with steam to avoid deterioration in quality. Do not use gamma rays to avoid colour changes of the product.